Table of Contents
Banner Engineering DBQ5 Proximity Sensor Tester for DC Sensors (Banner Engineering DBQ5 Portable Demo Box)
In the old days, my testing of dc proximity sensors was performed with my Fluke DMM. Not a problem for me but somewhat confusing to untrained and inexperienced team members. Trying to get them to understand PNP and NPN compounded the problem. One day, I browsed the McMaster Carr catalog, and I saw something new on the three-wire dc proximity sensors pages.
This new thing was called a “Proximity Sensor Tester.” Sounding too good to be true, I ordered a unit. The sensor unit was simple to use. Simply place the brown wire under the +24vdc terminal, the blue wire under the 24vdc terminal, and the remaining black wire under the sensor output terminal. A quick test would tell me if the sensor was normally open, normally closed, and whether it was PNP or NPN type output. Much easier than my Fluke DMM, the tester became a critical tool in my test tool arsenal.
Why the Banner DBQ5 Proximity Sensor Tester Demo Box is Better Than My Previous Sensor Tester
A few years later, my local Banner Engineering representative showed up with the “Banner Engineering DBQ5 Portable Demo Box” in his pocket. While the Banner DBQ5 didn’t do anything, my old sensor tester couldn’t do it, but it did do a few things better. There were four critical features that I would consider improvements.
- Metal Construction – The DBQ5 is rugged, it’s metal cased, and the LED’s are protected from physical damage.
- M12 5-pin pigtail for direct sensor attachment. This connection is an additional connection method. The DBQ5 also features spring load terminals similar to my original sensor tester.
- Two Inputs – The DBQ5 has provisions for two sensor outputs versus my original unit’s single point.
- 27VDC versus 18VDC of my original unit. Batter life is much greater with three nine-volt batteries versus two.
Banner Engineering DBQ5 Portable Demo Box Not Just For Testing Sensors
My DBQ5 sees heavy use for setting up cylinder position sensors and proximity sensors without operating the machine or even having it powered on. I have a small collection of adapters and cables I carry with the DBQ5.
I also use the DBQ5 as a portable power supply for setting up and testing photoelectric sensors while installed on a machine.
The Banner Engineering DBQ5 Portable Demo box will, for the foreseeable future, have a place in my toolbox as well the tool toolboxes of my team members. It makes the task of setting up and testing dc proximity sensors, cylinder position sensors, and similar devices simple and efficient. To use the DBQ5 requires a fraction of the skills a DMM would require for the same tasks.
Banner Engineering DBQ5 Portable Demo Box Description
- Used to power dc self-contained photoelectric sensors for testing purposes
- Designed around the 4-pin Euro-style connector
- Powered by three standard 9V batteries (27V dc) for very long service life
- Features bi-color LEDs that indicate not only sensor output status but also output type (NPN or PNP)
- Cable adapters are available to convert to pico-style or ministyle connectors.
- A 4-pin wiring barrier is mounted on the top of the box to allow the connection of cabled dc sensors.
Where to Buy The Banner DBQ5 Proximity Sensor Tester
Over the years, I have bought several Banner DBQ5 Proximity Sensor Testers for maintenance team members. I originally ordered them from my local Banner Engineering representative, but I found the delivery to be painfully slow. Now I buy them from Amazon Marketplace Sellers and have the tester in hand within days.
Additional Sensor Testers & Test Boxes
- Balluff Sensor Tester BAE TO-XE-012-01 (BAE00WH)
- Sick Test Box 6038940
- Turck Test Box TB3-CP80
- Baumer Digital Sensor Tester 11084377
- IFM Sensor Testpak E18401
- Festo Sensor Tester SM-TEST-1
- Pepperl+Fuchs Sensor tester ST 03
- Carlo Gavazzi ST03 Sensor Tester
- Banner Engineering DBRQ Rechargeable Demo Box
User Review( vote)